Input Capacitance : 3.0 PF +/- 0.5 PF Compensation Range : 5 PF~ 50 PF Cable Length: 2.0 Meter (+/- 0.2M) Temperature Cofficient: DC Accuracy: < = 0.7% Safety Approval : CE, GS, UL, CUL Operating Temperature: 0 ~ 50 Degree C. Storage Temperature : -20 ~ 70 Degree C/ Weight : 250 g Dimension: 75 Dia * 340 mm Length
-------------------------------------------- DP-25 Differential Probe 3 RANGE ATTENUATOR. 2 POWER SOURCE. 1000V INPUT CAT III LOW BATT. AUTO OFF. SEPARATING DESIGN.CONVENIENT, DURABLE. MEET CE, IEC 1010, EN 61010, UL 3111. Band width (-3dB, 50Ω load): DC-25MHz . ( x 20: DC~15MHz) Accuracy : ± 2% at 20-30°C 70% RH after 20 minutes warm up. Attenuation : x 20, x 50, x 200 ( Into 1M Ohm scope ) ; x 40, x 10 , x 400 ( Into 50 Ohm load ) Maximum operation Voltage ( DC + peak AC ) : Maximum input Differential Voltage : 1300V ( DC + peak AC ) or 1000V RMS Maximum input Voltage to Ground : 1000V ( DC + peak AC ) or 600V RMS Common Mode Rejection Ratio ( CMRR ): 60Hz : >10,000 : 1 100Hz: >1,000 : 1 1MHz: > 300 : 1 Noise : Input Impedance : 2M Ohm, 2.3 PF between inputs and ground. 4M Ohm, 1.2 PF between inputs. Power :One internal 9V alkaline battery or external 6V-9V DC.
Do not use DP-25 above 1000v ( DC + peak AC ) between ground and the input or 1,300V ( DC + peak AC ) between the input lead. Do not operate DP-25 in wet or damp condition. Do not operate DP-25 in an explosive atmosphere. Do not immerse DP-25 in liquids. Do not operated DP-25 without covers. Please change the battery when the "LOW BATT " LED is lighted. At this time DP-25 can operate but not guaranteed the accuracy. DP-25 can not operate if both POWER and LOW BATT LED are not light. FEATURES
The DP-25 FET input differential probe provieds a safe means of measuring circuits with floating potentials up to 1000 V ( DC+ peak AC ) from ground and 1300V ( DC + peak AC ) differential. The DP-25 converts the high voltage differencial input signal to a low voltage ground refereced signal for display on any Oscilloscope. The output BNC of DP-25 is calibrted to drive a high impedance ( 1M Ohm) load. INSTRUCTION FOR USE
Connect the output BNC of DP-25 to the input BNC of the Oscilloscope by the accessory BNC cable. Adjust the vertical offset of the Oscilloscope if necessary. Set the select proper range of the DP-25 and the V / DIV of the Oscilloscope according to the scale conversion chart. Scale conversion chart : The effective V / DIV is the attenuation factor of x 20, x 50, x 200 multiplied by the scale factor of the Oscilloscope. It will be twice when the 50 Ω load was used. For example, with the range set at x 200, and the scope set to 0.5 V / DIV, the effective V / DIV equals 200 x 0.5 or 100 V, with the 50 Ohm load was used, it becomes 200V, the power consumption will increase too. NOTE:If the voltage of the input signal exceeeds the linear range of the setting range.The signal output of the DP-25 would not accurately, the wave form display will be cut off.
11F, NO. 182, DUNHUA SOUTH ROAD, SEC. 2, DA'AN DISTRICT 10669 TAIPEI, TAIWAN (R.O.C.)